•Type: Magnetic Level Gauge
•Mounting Orientation: Top Mounted, Side Mounted & Top – Bottom
•Indicator System: Bi-colour Rotating Roller / Capsule
•Housing for Indicator: For Bi-colour Roller – RAIL in Aluminium or SS Housing System For
•Capsule – Heavy Walled Glass Tube in Aluminium / SS Housing
•MOC of Indicator: ABS Plastic / SS
•Float Chamber: Non-Magnetic Standard pipe
•MOC of Float Chamber: SS 304, SS 316, SS 316L, PP, PVDF, Others on request
•Float: SS 304, SS 316, SS 316L, PP, PVDF, Others on request
•Scale : Aluminium / SS / Acrylic engraved in mm
•Process Connection: Flanged in various sizes
•Vent: 1⁄2” Plugged / 1⁄2” Needle Valve / 1⁄2” Ball Valve Working Of Magnetic Level Indicator:
•Magnetic Level Indicator operates on the principle of magnetic field coupling to provide fluid level information. Float chamber is typically constructed with non-magnetic pipe having process connections that matches to the vessel connections.
•Float size and weight is determined by the process fluid, pressure, temperature and the specific gravity of the process fluid. Float contains magnets to provide 360o magnetic flux field.